Displayed in 5 seconds.
Tiny Blood Sample
Requires only 0.75µL.
Easy to handle and prevents blood contact for excellent hygiene.
Start monitoring quickly with this simple design.
Enhanced size for improved grip, ensuring accuracy and precision.
How it works
1Insert the strip in the socket of the meter, which will do calibration automatically.
2Place blood sample against the sample entry of strip.
3After 5 seconds, the test result appears.
4Rotate and pull up the test strip.
|Test strip||ELSA strip|
|Coding||Coding not required|
|Enzyme / Measurement technology||GOD / Electrochemical sensor|
|Blood sample source||Capillary whole blood|
|Minimum sample volume||0.75 µL|
|HCT range||30 - 57%|
|Measuring range||0.6 - 33.3 mmol/L (10 - 600 mg/dL)|
|Meter dimensions (L x W x H)||95.0 x 43.8 x 13.0 mm (3.74 x 1.72 x 0.51 inch)|
|Display area (L x W)||38.0 x 29.0 mm (1.49 x 1.14 inch)|
|Weight (with battery)||43.0 ± 5 g|
|Power supply||1 CR2032 battery|
|Memory capacity||500 test results with date and time|
|Operating temperature||10 - 40℃ (50 - 104°F)|
|Operating relative humidity||10 - 90%|
|Meter storage conditions||-10 - 60°C (14 - 140°F)|
|Test strip storage conditions||At 4 - 30°C (39 - 86°F), and below 90% relative humidity|
|Accuracy||For glucose concentrations < 5.5 mmol/L (100 mg/dL),
95% of measured glucose values within ± 0.83 mmol/L (15 mg/dL) of reference measurements.
For glucose concentrations ≥ 5.5 mmol/L (100 mg/dL),
95% of measured glucose values within ± 15% of reference measurements.
|Precision||For glucose concentrations < 5.5 mmol/L (100 mg/dL), SD < 0.28 mmol/L (5 mg/dL)
For glucose concentrations ≥ 5.5 mmol/L (100 mg/dL), CV < 5%
Fully comply with ISO 15197:2013 (For other received credentials, please contact us.)