Record and compare pre- and post-prandial results to understand how food affects your blood glucose.
Easy, wireless data transfer.
Easy to handle and prevents blood contact for excellent hygiene.
Eliminates the need for manual coding and delivers maximum accuracy.
Utilizes gold electrodes and a patented strip structure to provide accurate and reliable results.
Sync with Rightest CARE App
Glucose Management Made Simple
How it works
1Insert the strip into the socket of the meter. Calibration will occur automatically.
2Touch a blood sample against the sample entry port of the strip.
3After 5 seconds, the test result will appear.
4Rotate and pull upwards to remove the test strip.
|Enzyme / Measurement technology||GOD / Electrochemical sensor|
|Blood sample source||Capillary whole blood (finger, palm and forearm)|
|Minimum sample volume||0.75 µL|
|HCT range||20 - 60%|
|Measuring range||0.6 - 33.3 mmol/L (10 - 600 mg/dL)|
|Meter dimensions (L x W x H)||85.0 x 57.0 x 22.5 mm|
|Display area (L x W)||39.0 x 39.5 mm|
|Weight (with battery)||81.0 g|
|Power supply||2 × AAA batteries|
|Memory capacity||500 blood glucose test results with date and time|
|Operating temperature||10 - 40°C (50 - 104°F)|
|Operating relative humidity||10 - 90%|
|Meter storage conditions||-10 - 60°C (14 - 140°F)|
|Test strip storage conditions||4 - 30°C (39 - 86°F), <90% relative humidity|
|Accuracy||95% of the measured glucose values shall fall within either
Glucose concentrations < 5.5 mmol/L (100 mg/dL), difference within ± 0.83 mmol/L (15 mg/dL)
Glucose concentrations ≥ 5.5 mmol/L (100 mg/dL), bias within ± 15%
|Precision||Glucose concentrations < 5.5 mmol/L (100 mg/dL), SD < 0.28 mmol/L (5 mg/dL)
Glucose concentrations ≥ 5.5 mmol/L (100 mg/dL), CV < 5%
Fully compliant with ISO 15197:2013. (For other received credentials, please contact us.)