Bluetooth transmission (For Max Plus)
Transfer test results to smartphones wirelessly.
Wide HCT range
Measures glucose value of extreme hematocrit range, like neonate, elder, anemia patient.
Multiple blood sample
Capillary, venous, arterial and neonatal whole blood are applicable.
Eliminates the needs for manual coding and delivers better accuracy than no coding.
Intuitive markers for time status, exercise and ‘before’ or ‘after’ a meal.
Understand ketone notice at a glance.
- Wake up
- Before meal
- After meal
With Rightest Bluetooth glucometer
Manage blood glucose and maintain health easily
How it works
1Insert the strip in the socket of the meter, which will do calibration automatically.
2Place blood sample against the sample entry of strip.
3After 5 seconds, the test result appears.
4Rotate and pull up the test strip.
Max Series Specifications
|Test strip||Max Strip|
|Enzyme / Measurement technology||GDH / Electrochemical sensor|
|Blood sample source||Capillary, venous, arterial and neonatal whole blood|
|Minimum sample volume||0.75 microliters (0.00075c.c.)|
|HCT range||10 - 70%|
|Measuring range||0.6 - 33.3mmol/L (10 - 600mg/dL)|
|Meter dimensions (L x W x H)||82.0 x 50.0 x 15.5 mm (3.23 x 1.97 x 0.61 inch)|
|Display area (L x W)||40.7 x 40.2 mm (1.60 x 1.58 inch)|
|Weight (with battery)||59.0 ± 5 g|
|Power supply||1 CR2032 battery (Max) ｜ 2 CR2032 battery (Max Plus )|
|Memory capacity||500 blood glucose test results with date and time|
|Operating temperature||6 - 44°C (43 - 111°F)|
|Operating relative humidity||10 - 90%|
|Meter storage conditions||-10 - 60°C (14 - 140°F)|
|Test strip storage conditions||4 - 30°C (39 - 86°F), < 90% relative humidity|
|Bluetooth version (For Max Plus)||4.0|
|Accuracy||95% of the measured glucose values shall fall within either
Glucose concentrations < 5.5mmol/L (100mg/dL), difference within ± 0.83mmol/L (15mg/dL)
Glucose concentrations ≥ 5.5mmol/L (100mg/dL), bias within ± 15%
|Precision||Glucose concentrations < 5.5mmol/L (100mg/dL), SD < 0.28mmol/L (5mg/dL)
Glucose concentrations ≥ 5.5mmol/L (100mg/dL), CV < 5%
Fully comply with ISO 15197:2013 (For other received credentials, please contact us.)
- Getting Started Guide
- User's Manual
- Test Strip Guide
- Instruction for Lancing Device