Before-and-after meal markers help you know how food influences your blood glucose.
Transfer test results to smartphones wirelessly.
Blood contact prevention
Larger, thicker test strips help with dexterity issues. The side-insert design ensures better hygiene.
Eliminates the needs for manual coding and delivers better accuracy than no coding.
Utilizes gold electrodes and patented strip structure to provide accurate and reliable results.
With Rightest Bluetooth Glucose Meter
Manage blood glucose and maintain health easily
How it works
1Friendly - large strip hand bar
2Place blood sample against the sample entry of strip.
3Rapidly - results in 8 seconds
4Safety - keep from touching blood sample
GE 333 Specifications
|Test strip||GE 333|
|Enzyme / Measurement technology||GOD / Electrochemical sensor|
|Blood sample source||Capillary whole blood (finger, palm and forearm)|
|Minimum sample volume||0.75 microliters|
|HCT range||20 - 60%|
|Measuring range||0.6 - 33.3mmol/L (10 - 600mg/dL)|
|Meter dimensions (L x W x H)||85.0 x 57.0 x 22.5 mm|
|Display area (L x W)||39.0 x 39.5 mm|
|Weight (with battery)||81.0g|
|Power supply||2 AAA batteries|
|Memory capacity||500 blood glucose test results with date and time|
|Operating temperature||10 - 40°C (50 - 104°F)|
|Operating relative humidity||10 - 90%|
|Meter storage conditions||-10 - 60°C (14 - 140°F)|
|Test strip storage conditions||4 - 30°C (39 - 86°F), <90% relative humidity|
|Accuracy||95% of the measured glucose values shall fall within either
Glucose concentrations < 5.5mmol/L (100mg/dL), difference within ± 0.83mmol/L (15mg/dL)
Glucose concentrations ≥ 5.5mmol/L (100mg/dL), bias within ± 15%
|Precision||Glucose concentrations < 5.5mmol/L (100mg/dL), SD < 0.28mmol/L (5mg/dL)
Glucose concentrations ≥ 5.5mmol/L (100mg/dL), CV < 5%
Fully comply with ISO 15197:2013 (For other received credentials, please contact us.)
- Instructions for use
- Test Strip Guide
- Instruction for Lancing Device