Before- and after-meal markers help you know how food influences your blood glucose.
Helpful reminder for the next blood glucose measurement.
Easily use in dark environment.
Eliminates the needs for manual coding and delivers better accuracy than no coding.
Tiny blood sample required
Only 0.75 microliters (0.00075c.c.) of blood sample required.
Wide HCT range
Measures glucose value from hematocrit range including anemia, neonatal etc.
- Before meal
- After meal
Patented Rightest® lancing device GD720
How it works
1Insert a new disposable lancet firmly into lancet carrier and press the safety button.
2Press your finger smoothly on the cap.
3Apply a tiny blood sample.
How it works
1Take one test strip from the vial and close the cap immediately.
2Insert the test strip into the test strip port on the meter. The meter will automatically turn on.
3Press the security button and place the finger against the lancing device and press the finger on the cap.
4Gently squeeze your fingertip to get a drop of blood. Apply the blood drop on the edge of the test strip.
5After 5 seconds, the test result appears
6Rotate and pull up the test strip.
7Discard used test strip through the disposal channel.
|Accuracy||Glucose concentrations ＜ 100mg/dL (5.5mmol/L), error within ± 15mg/dL (0.83mmol/L)
Glucose concentrations ≧ 100mg/dL (5.5mmol/L), error within ± 15%
|Precision||Glucose concentrations ＜ 100mg/dL (5.5mmol/L), SD < 5mg/dL
Glucose concentrations ≧ 100mg/dL (5.5mmol/L), CV < 5%
|Coding||Auto coding calibration|
|Enzyme / Measurement technology||GDH / Electrochemical sensor|
|Blood sample source||Capillary, venous, arterial and neonatal whole blood|
|Minimum sample volume||0.75 microliters (0.00075c.c.)|
|HCT range||Glucose concentrations ≦ 200mg/dL, HCT range 20 - 70%
Glucose concentrations > 200mg/dL, HCT range 20 - 60%
|Measuring range||10 - 600mg/dL (0.6 - 33.3 mmol/L)|
|Meter dimensions (L x W x H)||71.3 x 39.0 x 14.0 mm (2.79 x 1.53 x 0.55 inch)|
|Display area (L x W)||32.7 x 25.02 mm (0.98 x 1.28 inch)|
|Weight (with battery)||50.0 ± 5 g|
|Power supply||2 CR2032 batteries|
|Memory capacity||1,000 blood glucose test results with date and time|
|Operating temperature||6 - 44℃ (43 - 111°F)|
|Operating relative humidity||10 - 90%|
|Meter storage conditions||-10 - 60°C (14 - 140°F)|
|Test strip storage conditions||4 - 30°C (39 - 86°F), < 90% relative humidity|
Fully comply with ISO 15197:2013 and FDA approval (For other received credentials, please contact us.)